Hair removal 808nm diode laser Beauty equipment
HR Hair removal
It is a professional model. In this model, you first choose patient skin color, sexual and treatment body part in Body Page(left picture above). Press “OK” to enter next page. Or press “reset” to rechoose.
Our smart system will give you recommended parameter of laser power, pulse width and frequency depending on your choice of sexual, skin color type and treatment body part. Those default will display in the corresponding column. If you want to follow those default, just press “OK”. If want to adjust your favourite parameter, press”SET”, then you can adjust laser power and frequency by “+” and “-”. Bottom “use recommended parameter” will reset those three parameter to default.
There are totally 3 parameters to be set, including: Laser Power, Frequency and Skin Cooling Level.
1) The Fluence shows the energy density of one laser shot. Laser Power fluence can be variable from 1J~120J. The step width is 1J. You can adjust it by press “+” and “-”.
2) The Frequency shows the pulses per second of the laser. A higher repetition rate leads to higher pulse energy fluence. The frequency can be variable from 1Hz~10Hz with a step width of 1Hz.
3) Subtotal at the top-right of the LCD screen shows the total laser energy of this treatment. It is easy for doctor to calculate how much power used in one treatment. Press button “clear” can clear the number. Or the subtotal number will clear automatically when machine shut down.
4) The Skin Cooling Level shows cooling efficiency of the Sapphire Window and Aluminum Tip of the handle piece. The green block is longer, the cooling is better. It depends on the feeling of the patents. The default cooling level is minimum output, which can be adjust by pressing “+” and “-” button. When the laser is working, you can not adjust skin cooling level. You may need to wait for several minutes before the skin cooling tip becomes cold.
5) The column below skin cooling shows the system status.
A. Status: Setting. This means you are setting parameters.
B. Status: Ready. When parameters have been set, press “READY” button, the system will enter ‘Ready’ status. In this status, the laser is ready to use.
C. Status: Working. When the handle piece shots laser pulses, the system will enter to ‘Working’ status.
D. Status: Water temperature high. The cooling water temperature range should be between 15~28℃. If it higher than 34℃, the laser will stop. Please wait till water temperature going down. The warning will disappear, and system can be set again.
E. Status: Water flow rate low. It means the cooling water flow rate lower than minimum requirement. If the water flow rate which show at the bottom of the screen is just low, not “0 L/min”. Please first check along the water tube to confirm whether there are any jam in it and handle piece water connector are well locked. Secondly, open the water inlet at the back of machine to make the air pressure balance between the inside and outside of water circle system. If the water flow rate show “0 L/min”. Please first check whether the water pump are working. Or contact our company to ask for help..
F. Status: Operation wrong. Press switch on the handle piece without enter ready status.
G. Status: Please check Diode laser or diode driver. It means the laser handle piece or laser power supply broken. Please ask factory or seller to solve this problem.
H. Status:Error 2. It means the control board have cooling problem. Please ask factory or seller to solve this problem.
6) Set the parameters (Fluence, Frequency, Cooling) and press ‘READY’ button to enable the handle piece. For the typical treatment parameters, you can refer to the recommended default or experienced doctor’s advice.
7) During ‘Ready’ status, you also can adjust the skin cooling depending on skin feeling. You may need to wait for several minutes before the skin cooling tip becomes cold.
8) Take the handle piece. Pressure the pedal switch or the switch on the handle piece to shot the laser pulses. Keep pressing the button and the laser output will continue. Do not press & release for single pulse.
9) You can stop the machine anytime by turn off the key in the front panel or press emergency button.
Treatment Notice of 808nm diode laser Hair removal Beauty equipment
1. The treatment of the hair removal needs experienced professionals. Do not use it if you have no experience for such kind of laser device.
2. Operators and patients must wear laser goggles to protect eyes from any laser
3. Please notice that this is a high power laser device and if the patients feel very
painful, please stop the laser and check if there is anything happens to the patients’ skin. Adjust it as a proper power that the patients can stand to do the treatment.
4. Skin color is very important. For white to light color, you can adjust the laser
power up to 25~40J/cm2. For light color to light brown, you need a lower laser
power (10~15J/cm2 for example), Also you need to use lower than 10J/cm2 and
extend the treatment time for the dark skin to inject enough energy to the skin and follicles.
5. The skin must be shaved before treatment.
6. The handle piece is better to be operated at SCANNING mode. It is faster for
7. Skin contact gel is no necessary for treatment. It won’t make the treatment better or
improve the skin feeling.
8. You need to wait for several minutes before the cooling tip becomes cold.
9. Keep contact the cooled Sapphire with skin. And you can also use the edge of the Aluminum
10. head of the handle piece to cool down the skin during treatment.
11. Never shot the laser pulses to eyes, or watch the laser directly.
Advantage of 808nm diode laser Hair removal Beauty equipment
Safety: 808nm diode laser Hair removal Beauty equipment has stable performance, long usage life: it is real time controlled by the intellectual CPU.
Fast: Fast coverage rate for hair removal with large 11×11 mm square spot, high efficiency,
Efficacy: 808nm wavelength laser, which is in the spectrum of near infrared (0.75-1.50μm), could be absorbed by the melanin best, penetrates through the depth of dermis and adipose tissue and then act on the hairs with different positions and depths.
Pain free: Sapphire cooling tip, cools the epidermis down to 0~4 °C. Offers cooling all they way, comfort and without pain.
Convenience: Friendly use design, easy to operate man machine interface– touch display, easy to operate.Permanence hair removal: Applicable to all skin color, excellent result.
|Skin Cooling temperature||-5°C–5°C by cooled Sapphire|
|System Cooling method||Freon compressor|
|Input VAC||220VAC / 50-60Hz|
|Weight of machine||35KG|